FABOOL Laser CO2(Laser Tube with Laser Pointer) Optical Axis Adjustment Manual
Attach reflector 2 to the mirror mount.
Irradiate reflector 3.
Start from the same distance as when irradiating mirror 2.
Moving on, move the X-axis to the center of the processing area.
Adjust mirror mount 2 according to the amount of deviation. If there is significant vertical deviation, adjust the aluminum frame itself that holds mirror 2 little by little as shown in the picture. Fine tune it using the adjustment screw on the mirror mount. Make minor adjustments according to the deviation of the laser from close and far distances.
If there is significant horizontal deviation, adjust by turning mirror mount 2 little by little as show in the picture.
Fine tune using the mirror mount adjustment screws. Make minor adjustments according to the deviation of the laser from close and far distances.
After adjusting, ensure that the center of reflector 3 is irradiated.
Attach reflector 3. Fine tune using the mirror mount adjustment screws. After adjusting, ensure that the center of the laser head part is irradiated.
If the laser beam deviates, adjust the adjustment screws. Adjust the inclination of the mirror mount screws if there is significant horizontal deviation.
After adjusting, ensure that the center of the laser head part is irradiated.
Radiate the laser from all four corners of the processing area and ensure that the center of the laser head is irradiated.
Left Front of Processing Area
Ride Front of Processing Area
Left Back or Processing Area
Right Back of Processing Area
Attach the condenser lens.
*Be aware of the orientation of the condenser lens. The convex side is the top.
After attaching condenser lens, optical axis adjustment is complete.